KEITHLEY 196 MANUAL PDF
December 28, 2019 | by admin
Visit us on the web at for more information: • Price Quotations • Drivers· Technical Specifications. Manuals and Documentation . Model System DMM Instruction Manual Rev. D. Manual Type: Primary User; Part Number: D. Release Date: Sep 17 View and Download Keithley DMM instruction manual online. 5 function autoranging digital multimeter. DMM Measuring Instruments pdf manual.
|Published (Last):||3 November 2013|
|PDF File Size:||18.4 Mb|
|ePub File Size:||12.90 Mb|
|Price:||Free* [*Free Regsitration Required]|
ABORT keihley 7 3. Therefore, maintain constant temperatures to minimize these thermal emfs. Offset-compensation is selected through front panel Program 0 see paragraph 2.
To do so, simply add up the decimal bit values for the required SRQ conditions. It helps to shield resistance greater than lOOkfl to achieve a stable reading.
Keithley System – In Stock, We Buy Sell Repair, Price Quote
The instrument will power up to these default conditions. The instrument will return to the previously selected function and range. User saved defaults will be lost since factory default conditions will be saved.
The informa- tion is divided into the following sections: There is no indication on the display when cal mode has been enabled. Repeat the mV measurement at the other frequencies specified in Table LSN will be off when the unit is mwnual the listener idle state.
Keithley 196 System Digital Multimeter
Filter windows for the internal filter function in the same manner as the windows for the front panel filter. The 0 indicator light reveals the status of ohms offset compensation.
A filter value of 10 is recommended. The most recent defined word will be listed first.
Digital Calibration— When performing digital calibration, two points must be calibrated on each range. The U3 command allows the user to ieithley out the current defined size of the buffer.
System DMM Instruction Manual | Keithley – TechRecovery, LLC
However, since the instrument cannot make resistance measurements F2 at the selected interval QlOOshort period errors will occur. The following instrument operating parameters are saved by this program: The Q command is in the following form: Msnual Measure unit instruments Tech: Depressing and releasing the switch once turns the power on.
Exploring the boundaries of materials science or device development? Earlier ver- sions had different screws, which were silver colored. To get the instrument up and rurming quickly use the following procedure. FOX— Single command string.
December 3, at 2: The zero correc- tion procedure can be found in paragraph 2. Some of the statements have two forms, with the exact con- figuration depending on the command to be sent over the bus. Firmware Version With this program, the Model will indicate whether or not a specific reading falls within a prescribed range.
196 System DMM Instruction Manual | Keithley
According to the service manual, you should only flash the same version of the firmware as what is currently on your instrument. Commands sent without the execute character will not be executed at that time, but they will be retained within an internal command buffer for execu- tion at the time the X character is received.
Zero Correction— The Model must kithley properly zeroed when using the mV DC or the range in order to achieve rated accuracy specifications. K command options include: Repeat the procedure for each of the ranges kwithley negative voltages.