ASTM E317-11 PDF
December 21, 2019 | by admin
ASTM E Standard Practice for Evaluating Performance Characteristics of Ultrasonic Pulse- Echo Testing Instruments and Systems without the Use of. ASTM E – Standard Practice for Evaluating Performance Characteristics of Ultrasonic Pulse-Echo Testing Instruments and Systems without the Use of. Re: Need ASTM E Urgent please help. Here. Attached Files. File Type: pdf E STD Practice for Evaluating Performance Characteristics of UT.
|Published (Last):||5 October 2009|
|PDF File Size:||10.51 Mb|
|ePub File Size:||6.69 Mb|
|Price:||Free* [*Free Regsitration Required]|
Enter your account email address to request a password reset: Remember me for one month. Certain sections are applicable only to instruments having receiver gain controls calibrated in decibels dB.
Substitution of these methods is not precluded where appropriate; however, their usage is not within the scope of this practice. Today, some 12, ASTM standards are used around the world to improve product quality, enhance safety, facilitate atsm access and trade, and build consumer confidence.
Referenced Documents purchase separately The documents listed below are referenced within the subject standard but are not provided as part of the standard.
No items in cart. You need to log in before adding standards to the monitoring service. Please login to your authorized staff account to use this feature. A web subscription provides an easy and secure access to standards, and you are guaranteed to always have the latest edition. This is the prerogative of the using parties and is dependent on application, environment, and stability of equipment. This ensures that you are always up-to-date and saves you both time and money.
FORMAT Single User Subscription Site License Subscription BOOKS All Books PACKAGES All Packages EDITION to PUBLISHER. Webprint Printed and bound. E Standard Terminology Relating to Metallography. Significance and Use This practice describes procedures applicable to both shop and field conditions. Are you sure you want to empty the cart?
The validity of the results obtained will depend on the precision of the instrument display readings. Historical Version s – view previous versions of standard Translated Version s: Register for a trial account.
Standards monitoring is a free service designed for you who wish to be alerted of changes to or new editions of the standards required in your work. Implementation may require more detailed procedural instructions in the format of the using facility. Their use, however, should be mutually agreed upon and so identified in the test report.
The moment new editions of the standards you monitor are released, changes are made, or appendixes are published, you will be alerted by email. Active view current version of standard. E Standard Terminology for Nondestructive Examinations.
E371-11 to Active This link will always route to the current Active version of the standard. This document does not establish system acceptance limits, nor is it intended as a comprehensive equipment specification.
This is assumed to be. There are limitations to the use of standard reference blocks for that purpose.
While these may sometimes be designated gain, attenuator, or sensitivity on various instruments, the term gain controls will be used in this practice in referring to those which specifically control instrument receiver gain but not including asmt, electronic distance-amplitude compensation, or automatic gain control. This is the prerogative of the using parties and is dependent on application, environment, and stability of equipment.
Substitution of these methods is not precluded where appropriate; however, their usage is not within the scope of this practice. Where acceptance criteria are implied herein they are for example only and are subject to more or less restrictive limits imposed by customer’s and end user’s controlling documents.
The values given in parentheses are mathematical conversions to SI units that are provided for information only and are not considered standard.
No frequency of system evaluation or calibration is recommended or implied. Click here to download full list of books. These procedures can generally be applied to any combination of instrument and search unit of the commonly used types and frequencies, and to most straight-beam examination, either contact or immersed.
This is assumed to be The following editions for this book are also available There are limitations to the use of standard reference blocks for that purpose. Where acceptance criteria are implied herein they are for example only and are subject to more or less restrictive limits imposed by customer’s and end user’s controlling documents. In the case of evaluation of a complete system selection of the specific tests to be made should be done cautiously; if the related parameters are not critical in the intended application, then their inclusion may be unjustified.
Historical Version s – view previous versions of standard. More comprehensive or precise measurements of the characteristics of complete systems and their components will generally require laboratory techniques and electronic equipment such as oscilloscopes and signal generators. Instrument characteristics measured in accordance with this practice xstm expressed in terms that relate to their potential usefulness for ultrasonic testing.
Instrument characteristics expressed in purely electronic terms may be measured as described in E There were no books found astk the applied search filters. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.